Item no. HI-98703
Turbidity Meter with Fast Tracker™ Technology, EPA Compliant
High accuracy at low ranges (below 0.05 NTU)
Two, three or four point calibration
USB and RS232 PC connectivity
Log up to 200 records
User friendly, backlit display with guidance codes
Battery % on startup
Continuous current time on display
The HI 98703 meets and exceeds the requirements of the USEPA Method 180.1 for wastewater and Standard Method 2130 B for drinking water. The instrument has an EPA compliance reading mode which rounds readings to meet EPA reporting requirements. Users will appreciate the accuracy and sensitivity of this instrument, particularly at very low turbidity levels.
This instrument incorporates complete GLP (Good Laboratory Practice) functions that allow traceability of the calibration conditions. The last calibration points, time and date can be checked at the touch of a button.
With it’s logging function, up to 200 measurements along with it’s tagged locations can be stored in internal memory and consulted at any time. Data can be later transferred to a PC via RS232 or USB interface and HANNA HI 92000 software (optional).
Exclusive Fast Tracker™
For advanced field applications, the HI 98703 is equipped with Fast Tracker™—Tag Identification System (T.I.S.) that makes data collecting and management simpler than ever. Fast Tracker™ allows users to record the time and location of a specific measurement or series of measurements.
HI 98703-02 (230V) and HI 98703-03 (AUS plug) are supplied with sample cuvettes and caps (5), HI 98703-11 calibration cuvettes, HI 93703-58 silicone oil, cuvette cleaning cloth, batteries, AC adapter, instruction manual and rugged carrying case.
|Accuracy||±2% of reading plus 0.02 NTU|
|Calibration||Two, three or four-point calibration|
|Display||60 x 90 mm LCD with back light|
|Environment||To 50°C (122°F); max|
|Lamp life||Greater than 100,000 readings|
|Light Detection||silicon photocell|
|Light Source||Tungsten filament lamp|
|Logging Memory||200 records|
|Measuring mode||Normal, Average, Continuous|
|Method||Ratio Nephelometric Method (90°), ratio of scattered and transmitted light; Adaptation of the USEPA Method 180.1 and Standard Method 2130 B.|
|Range||0.00 to 9.99; 10.0 to 99.9 and 100 to 1000 NTU|
|Repeatability||±1% of reading or 0.02 NTU, whichever is greater|
|Resolution||0.01 NTU from 0.00 to 9.99 NTU; 0.1 NTU from 10.0 to 99.9 NTU; 1 NTU from 100 to 1000 NTU|
|Serial Interface||USB 1.1 or RS 232|
|Stray Light||< 0.02 NTU|
|Turbidity Standards||1, 10, 100 and 750 NTU|
|Typical EMC Deviation||±0.05 NTU|