Item no. HI-88703-02
Precision Turbidity Benchtop Meter, EPA Compliant
The HI88703 is a high accuracy (±2% of reading plus 0.02 NTU) Turbidity Benchtop Meter. The meter is supplied complete with AMCO-AEPA-1 primary turbidity standards used for calibration and performance verification. The HI88703 meets and exceeds the requirements of EPA Method 180.1 and Standard Methods for the Examination of Water and Wastewater 2130 B for turbidity measurements.
The HI88703 Precision Turbidity Benchtop Meter is specially designed for water quality measurements, providing reliable and accurate readings, especially in the low turbidity range. The instrument is based on a state-of-the-art optical system which guarantees accurate results, assures long term stability, and minimizes stray light and color interferences. Periodic calibration with the supplied standards compensates for any variations in intensity of the tungsten lamp. The 25 mm round cuvettes composed of special optical glass guarantee the repeatability of turbidity measurements.
Two Measurement Modes - The HI88703 features two options for turbidity measurement: ratio and non-ratio mode. Turbidity measurements can be made in the 0.00 to 4000 NTU (Nephelometric Turbidity Units) when using the ratio mode and in the 0.00 to 40.0 NTU range when non-ratio mode is used.
Multiple Turbidity Units of Measure - Turbidity can be read as Nephelometric Turbidity Units (NTU), European Brewing Convention units (EBC), or Nephelos units.
EPA Compliant - The HI88703 meets and exceeds the requirements of EPA and Standard Methods for turbidity measurements. When the meter is in EPA mode all turbidity readings are rounded accordingly to meet reporting requirements.
Calibration - The HI88703 has a powerful calibration function that compensates for lamp aging or changing. The calibration can be done using the supplied calibration solutions or user-prepared standards. A two, three, four, or five-point turbidity calibration can be performed by using the supplied (<0.10, 15.0, 100, 750, and 2000 NTU) standards. Calibration points can be modified if user-prepared standards are used.
AMCO AEPA-1 Primary Turbidity Standard - The AMCO AEPA-1 supplied standards are recognized as a primary standard by the USEPA. These non-toxic standards are made of styrene divinylbenzene polymer spheres that are uniform in size and density. The standards are reusable and stable with a long shelf life.
GLP Data - The HI88703 features complete GLP (Good Laboratory Practice) functions that allow traceability of the calibration conditions. Data includes calibration points, date, and time.
Data Logging - Up to 200 measurements can be stored in the internal memory and recalled at any time.
Data Transfer - For further storage or analysis options, logged data can be downloaded to a Windows compatible PC using the USB port and the HI92000 software.
Tutorial Mode - The unique tutorial mode provides additional information to help the user during measurements. When enabled, the instrument displays explanations and a confirmation button when a preparation or other operation has to be performed.
Contextual Help - Contextual help is always available through a dedicated HELP button. Clear tutorial messages and directions are available on-screen to quickly and easily guide users through setup and calibration. The help information displayed is relative to the setting/option being viewed.
Backlit Graphic LCD Display - A graphic LCD display provides an easy to understand, user-friendly interface. All messages are in plain text making them easy to read.
Turbidity is one of the most important parameters used to determine the quality of drinking water. Once considered as a mostly aesthetic characteristic of drinking water, significant evidence exists that controlling turbidity is a competent safeguard against pathogens. In natural water, turbidity measurements are taken to gauge general water quality and its compatibility in applications involving aquatic organisms. The monitoring and treatment or wastewater was once solely based on the control of turbidity. Currently, the measurement of turbidity at the end of the wastewater treatment process is necessary to verify that the values are within regulatory standards.
Turbidity of water is an optical property that causes light to be scattered and absorbed, rather than transmitted. The scattering of light that passes through a liquid is primarily caused by the suspended solids present. The higher the turbidity, the greater the amount of scattered light. Even a very pure liquid will scatter light to a certain degree, as no solution will have zero turbidity.
Drinking water treatment plants that obtain water from surface water are required by the EPA to monitor and report the turbidity. Surface water sources include lakes and rivers. The requirements of the nephelometer and reporting measurements according to EPA Method 180.1 are:
Report results as follows:
|NTU Reading||Rounded to nearest|
|0.0 - 1.0||0.05|
|1 - 10||0.1|
|10 - 40||1|
|40 - 100||5|
|100 - 400||10|
|400 - 1000||50|
The HI88703 meets and exceeds the meter criteria specified by the EPA Method 180.1 and Standard Methods for the Examination of Water and Wastewater 2130 B.
Order Information: HI 88703-02 (230V) is supplied with sample cuvettes and caps (5), calibration cuvettes, silicone oil (HI 93703-58), tissue for wiping cuvettes, power cord and instruction manual.
|NTU ratio Range||0.00 to 9.99; 10.0 to 99.9; 100 to 4000 NTU
0.0 to 99.9; 100 to 26800 Nephelos
0.00 to 9.99; 10.0 to 99.9; 100 to 980 EBC
|NTU ratio Resolution||0.01; 0.1; 1 NTU
0.1; 1 Nephelos
0.01; 0.1, 1 EBC
|NTU non ratio Range||0.00 to 9.99; 10.0 to 40.0 NTU 0.0 to 99.9; 100 to 268 Nephelos 0.00 to 9.80 EBC|
|NTU non ratio Resolution||0.01; 0.1 NTU 0.1; 1 Nephelos 0.01 EBC|
|NTU Accuracy||±2% of reading plus 0.02 NTU (0.15 Nephelos; 0.01 EBC)
±5% of reading above 1000 NTU (6700 Nephelos; 245 EBC)
|Turbidity Repeatability||±1% of reading or 0.02 NTU (0.15 Nephelos; 0.01 EBC) whichever is greater|
|Stray Light||< 0.02 NTU (0.15 Nephelos; 0.01 EBC)|
|Turbidity Light Detector||Silicon Photoce|
|Turbidity Method||Nephelometric method (90°) or Ratio Nephelometric Method (90° & 180°)
Adaptation of the USEPA Method 180.1 and Standard Method 2130 B.
|Measuring mode||Normal, Average, Continuous|
|Turbidity Standards||<0.1, 15, 100, 750 and 2000 NTU|
|Calibration||Two, three, four or five-point calibration|
|Light Source||Tungsten filament lamp|
|Lamp Life||greater than 100,000 readings|
|Display||40 x 70 mm graphic LCD (64 x 128 pixels) with backlight|
|LOG Memory||200 records|
|Environment||0 °C to 50 °C; max 95% RH non-condensing|
|Power Supply||230 V/50 Hz|
|Auto Shut-Off||After 15 minutes of non-use|
|Dimensions||230 x 200 x 145 mm L x W x H|